Radar level measurement wholesale with kaidi86.com? The electromagnetic wave emitted during measurement can pass through the vacuum, no transmission medium is required, and the anti-interference ability is strong, and it is not affected by temperature, wind, water vapor, water mist, rain, etc. It can be used for liquid level measurement of almost all liquids, and can be continuously measured on-line unattended. Large measurement range, high precision, mature technology and stable performance. Read even more information at radar level sensor.
Under many operating conditions, ultrasonic level meter and radar level meter are commonly used. Some users are very entangled in the choice of these two level meter and do not know how to choose. Today, let’s talk about the principles and selection principles of these two types of level meter . Principle and selection principle of ultrasonic level meter. Working principle: The ultrasonic pulse probe emits a beam of ultrasonic pulses to the measured medium, and the sound wave is reflected by the liquid surface. The distance between the liquid levels is measured by measuring the time difference between the emission and reflection of the sound waves. Since the ultrasonic level gauge is not a liquid, it can be used to measure corrosive, non-volatile and non-foaming places.
So what can be done about these difficulties? Under the condition of strong dust, on the one hand, the radar with high transmitting energy can be selected, on the other hand, the measurement software with continuous measurement algorithm of wave-loss waiting can be selected. When the radar encounters strong dust, it will not misjudge the measurement result even if the radar loses wave for a short time. After entering the state of continuous measurement algorithm, if the reflection wave of real material surface can be recognized within the set waiting time, the correct measurement value of material surface can be obtained. In the past, only a few foreign radars have this function. At present, there are also domestic radars with this function, and the practical application effect is very good.
If the radar level meter is unreasonably selected, the interference echo cannot be handled well, and the reliability of the instrument will be reduced. Therefore, the following factors should be considered when selecting a radar level meter: Conductivity and dielectric constant of the measured medium. The measured medium is a conductive liquid or a liquid with a dielectric constant above 4. Generally, a common radar is selected. Liquids with small dielectric constants (dielectric constants below 2) and some conductive solids often use precision radars or guided wave radars due to the large amount of interference echoes.
Thermal interference. During the operation of the thermal power plant, its thermal equipment will generate a lot of heat, causing changes in the surrounding instruments and ambient temperature. This is what we call thermal interference. These disturbances can affect the components of the magnetrol radar level gauge and further create problems such as inaccurate measurements. Light interference, light interference mainly exists in semiconductor components. Many components used to control instruments are made of semiconductor materials whose conductivity changes under the influence of light. This will affect the normal use of the radar level gauge.
As one of the most professional magnetic level indicators manufacturers in China,Guangdong Kaidi Energy Technology Co., Ltd. provides customized solutions for a range of industrial automation process applications,such as mechanical float level indicator.We specialized in radar level gauge, fork type level switch,etc. These were implemented successfully, and KAIDI magnetic level gauge manufacturers products,such as radar level meter, magnetic level indicators, can be used in many different industries such as food & beverage, water, energy, pharmaceutical etc. Read even more details on kaidi86.com. The Magnetic Level Gauge all use vacuum tube technology, with a lifespan of 3-5 years, and protection grade is up to IP68, not easy to fade.
Working principle: Tube in the liquid level meter has a magnetic float, float design, according to the proportion of medium float magnet steel and container liquid level at the same level, wholesale liquid level gauge in a semiconductor magnetel liquid level gauge sensor LED display scale, and orderly arranged from top to bottom, and LED module corresponding to the magnetic field magnetic sensor, the magnetic semiconductor, triggering the corresponding digital circuit, liquid level above the red LED display, The LED below the liquid level gauge is displayed in green, the red in gas phase, and the green in liquid phase. The red-green junction is the actual liquid level in the container.